Mô tả sản phẩm
Product image  Type  Description 

ASGA Smallsized, Smallcapacity Acceleration Transducer 
The ASGA is an oil damping type smallsized acceleration transducer for measuring small acceleration. Its cable can be removed easily. Rated Capacity: ±9.807 to ±196.1 m/s2 Dimensions: 14 x 18 x 14 mm Rated Output: 0.5 mV/V or more Frequency Response: DC to (40 to 250) Hz 

ASGB Smallsized, Smallcapacity Acceleration Transducer 
The ASGB is an oil damping type smallsized acceleration transducer for measuring small acceleration. Its cable fixed to the sensor that cannot be removed. Rated Capacity: ±9.807 to ±196.1 m/s2 Dimensions: 14 x 20 x 14 mm Rated Output: 0.5 mV/V or more Frequency Response: DC to (40 to 250) Hz 

ASB Smallsized Acceleration Transducer 
The ASHB is an oil damping type smallsized acceleration transducer with frequency response wider than ASB. Rated Capacity: ±98.07 to ±490.3 m/s2 Dimensions: 16 x 16.5 x 16 mm Rated Output: 0.5 mV/V ±20 (±25% for 10HB) Frequency Response: DC to (500 Hz to 1.5 kHz) 

ASHB Smallsized, High Frequency Response Acceleration Transducer 
The ASHB is an oil damping type smallsized acceleration transducer with frequency response wider than ASB. Rated Capacity: ±98.07 to ±490.3 m/s2 Dimensions: 16 x 16.5 x 16 mm Rated Output: 0.5 mV/V ± 20% (±25% for 10HB) Frequency Response: DC to (500 Hz to 1.5 kHz) 

ASHA Smallsized, High Frequency Response Acceleration Transducer 
The ASHA is an oil damping type smallsized acceleration transducer that can response high frequence. Rated Capacity: ±980.7 to ±9807 m/s2 Dimensions: 15.2 x 15.5 x 14 mm Rated Output: 0.5 mV/V ± 20% Frequency Response: DC to (2 to 7) kHz 

ASTG Smallsized Triaxial Acceleration Transducer 
The ASTG is an oil damping type smallsized triaxial acceleration transducer for synchronously measuring acceleration in 3 directions of back & forth, right & left, and up & down. Rated Capacity: ±9.807 to ±196.1 m/s2 Dimensions: 44 x 20 x 44 mm Rated Output: 0.5 mV/V Frequency Response: DC to (40 to 250) Hz 

ASTB Triaxial Acceleration Transducer 
The ASTB is an oil damping type smallsized triaxial acceleration transducer for synchronously measuring acceleration in 3 directions of back & forth, right & left, and up & down. Accurate measurement possible due to little interference among axes. Rated Capacity: ±98.07 to ±490.3 m/s2 Dimensions: 44 x 20 x 37 mm Rated Output: 0.5 mV/V or more Frequency Response: DC to (350 Hz to 1 kHz) 

ASWA Waterproof Acceleration Transducer 
The ASWA is an oil damping type waterproof acceleration transducer. It can withstand against water pressures up to 490 kPa. Rated Capacity: ±9.807 to ±196.1 m/s2 Dimensions: 18 x 18 x 24 mm Rated Output: 0.5 mV/V or more Frequency Response: DC to (40 to 250) Hz 

ASQD Servo Type Acceleration Transducer 
The ASQD is a servo type acceleration transducer for accurate measuring minute vibration. A dedicated signal conditioner VAQ is available as the mating instrument. Rated Capacity: ±9.807 to ±49.03 m/s2 Dimensions: 32.5 x 75 x 32.5 mm Rated Output: ±5V (10 V) ±5% Frequency Response: DC to 100 Hz 

ASPAA Piezoelectric Acceleration Transducer (Builtin Amplifier) 
The ASPAA is a piezoelectric acceleration transducer with builtin amplifier for measuring high range from slight to large acceleration in high sensitivity. Applicable acceleration: ±2200m/s2 Dimensions: φ7.8 x 15.7 x Hex7 mm Voltage Sensitivity: 1.0 mV per m/s2 ±10% Power Supply: 15 to 25 VDC, 0.5 to 5.0 mA 

ASPBA Piezoelectric Acceleration Transducer (Builtin Amplifier) 
The ASPBA is a piezoelectric acceleration transducer with builtin amplifier for measuring high range from slight to large acceleration in high sensitivity. Applicable acceleration: ±2200 m/s2 Dimensions: φ8.5 x 11.5 x Hex9 mm Voltage Sensitivity: 1.0 mV per m/s2 ±10% Power Supply: 15 to 25 VDC, 0.5 to 5.0 mA 

ASPCA Piezoelectric Acceleration Transducer (Builtin Amplifier) 
The ASPCA is a piezoelectric acceleration transducer with builtin amplifier for measuring high range from slight to large acceleration in high sensitivity. Applicable acceleration: ±400 to ±3600 m/s2 Dimensions: 14.2 x 14.2 x 23.7 mm Voltage Sensitivity: 1.0 to 10 mV per m/s2 ±10% Power Supply: 21to 24 VDC, 0.5 to 10 mA 

AMAA Smallsized Triaxial Accelerometer 
The AMAA is a compact & lightweight accelerometer with large overloads resistant. Rated Capacity: ±19.61 to ±490.3 m/s2 Dimensions: 10 x 5.8 x 14 mm (±19.61, ±49.03m/s2), 11 × 6.5 × 15 mm (±98.07m/s2), 11 × 11 × 19.3 mm (±490.3m/s2) Rated Output: ±2 V ±0.2 V Frequency Response: DC to 500 Hz (±19.61, ±49.03m/s2), DC to 200 Hz (±98.07, ±490.3m/s2) Safe Overloads: ±19613 m/s2 (±19.61, ±49.03m/s2), ±9807 m/s2 (±98.07, ±490.3m/s2) 

GSATA Three Axis Angular Rate Gyro 
The GSATA is a compact & lighweight angular rate gyro for measurig anglar rate of 3 axes. Rated Capacity: ±900 deg/s (±15.708 rad/s) Dimensions: 10 x 8 x 20 mm Rated Output: Approx. ±2 V/V Recommended Excitation: DC ±2.5 to 5.0 V with dual supply Safe Shock Resistant: 9807 m/s2 

ASDB Miniature Crash Test Acceleration Transducer 
The ASDB is a undamped type compact & lighweight acceleration transducer for crash tests. There are two types vertical models ASDB1KH/2KH and horizontal models ASDB1KV/2KV. Rated Capacity: ±9807 to ±19613 m/s2 Dimensions: 10 x 4.7 x 12.5 mm Rated Output: 1.2 mV/V or more Frequency Response: DC to 4 kHz, sensitivity deviation: ±5% at 23ºC 

ASDEA Damped Acceleration Transducer for Crash Test 
The ASDEA is an oildamped type acceleration transducer for crash tests. It is least affected from resonance. Rated Capacity: ±980.7 to ±9807 m/s2 Dimensions: 10 x 4.9 x 13.5 mm Rated Output: 1 mV/V or more Frequency Response: DC to 1 kHz, sensitivity deviation ±3%, 1 kHz to 4 kHz, sensitivity deviation ±10% at 23ºC 

ASDHA Miniature Crash Test Acceleration Transducer 
The ASDHA is a undamped type compact & lighweight acceleration transducer with a sputtertype strain gage as the sensing element. There are two types vertical (ASDHA1KH/2KH) and horizontal (ASDHA1KV/2KV). Rated Capacity: ±9807 to ±19613 m/s2 Dimensions: 10 x 4.7 x 12.5 mm Rated Output: 1.2mV/V or more Frequency Response: DC to 4 kHz, sensitivity deviation ±5% at 23ºC 

ASDRA Dumped Accelerometer for Automotive Crash Tests 
The ASDRA is an oil damped type acceleration transducer for crash tests with exchangeable cable. It is least affected from resonance. Rated Capacity: ±9807 m/s2 Dimensions: 5.4 x 5 x 13 mm Rated Output: 2 mV/V or more Frequency Response: DC to 1 kHz (Sensitivity deviation±3%), 1 k Hz to 4 kHz (Sensitivity deviation ±10% at 23ºC) 

ASEA Damped Acceleration Transducer for Crash Test 
The ASEA is an oil damped acceleration transducer for crash tests. It is suitable for measuring vibration of car bodies, components, and dummies. Rated Capacity: ±980.7 to ±9807 m/s2 Dimensions: 10 x 5 x 10 mm Rated Output: 0.5 mV/V or more (±980.7 m/s2), 1 mV/V or more (±4903 m/s2), 2 mV/V or more (±9807 m/s2) Frequency Response (23ºC): DC to 1 kHz for all models (Sensitivity deviation ±3%), 1 kHz to 2 kHz for ASEA100 (Sensitivity deviation ±10%) (±980.7 m/s2), 1 kHz to 4 kHz for ASEA500/1K (Sensitivity deviation ±10%) (±4903 m/s2) 

ASERA Damped Acceleration Transducer for Crash Test 
The ASERA is an oildamped type, suitable for measuring car bodies and dummies. Its cable is exchangeable by repair. Rated Capacity: ±980.7 to ±9807 m/s2 Dimensions: 10 x 5 x 10 mm Rated Output: 0.5 mV/V or more (±980.7 m/s2), 1 mV/V or more (±4903 m/s2), 2 mV/V or more (±9807 m/s2) Frequency Response: (23ºC): DC to 1 kHz for all models (Sensitivity deviation ±3%), 1 kHz to 2 kHz for ASERA (Sensitivity deviation ±10%) (±980.7 m/s2), DC to 1 kHz for ASERA500/1K (Sensitivity deviation ±3%) (±9807 m/s2) 

ASM1KBCH M3 Miniature Semiconductor Acceleration Transducer 
The ASM1KBCH M3 is an undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. The sensitivity direction is parallel to the mounting surface. Rated Capacity: ±9807 m/s2 Dimensions: 12 x 5.4 x 25 mm Rated Output: 1.75 mV/V or more Frequency Response: DC to 2.8 kHz, sensitivity deviation ±5% at 23ºC 

ASM1KBCV M3 Miniature Semiconductor Acceleration Transducer 
The ASM1KBCV M3 is an undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. The sensitivity direction is perpendicular to the mounting surface. Rated Capacity: ±9807 m/s2 Dimensions: 12 x 5.4 x 25 mm Rated Output: 1.75 mV/V or more Frequency Response: DC to 2.8 kHz, sensitivity deviation: ±5% at 23ºC 

ASM1KCBV Miniature Semiconductor Acceleration Transducer 
The ASM1KCB is a undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. The cable is exchangeable. Rated Capacity: ±9807 m/s2 Dimensions: 16 x 7.7 x 12.5 mm Rated Output: 1.5 mV/V or more Frequency Response: DC to 3.5 kHz, sensitivity deviation: ±5% at 23ºC 

ASM200BA Miniature Semiconductor Acceleration Transducer 
The ASM200BA is an undamped type semiconductor acceleration transducer for crash tests. Compact & lightweight ensure minimal influence to the vibration mode of a measuring object. Rated Capacity: 1961 m/s2 Dimensions: 7.6 x 4.6 x 15 mm Rated Output: 1.3 mV/V or more Frequency Response: DC to 3.5 kHz, sensitivity deviation: ±5% at 23ºC 

ASMAID Semiconductor Acceleration Transducer for Crash Test 
The ASMAID is a semiconductor acceleration transducer for crash tests. Rated Capacity: ±19614 m/s2 (2000 G) Rated Output: 6 mV/V or more Frequency Response: DC to 2.8 kHz, sensitivity deviation: ±5% at 23ºC 
Tags: Acceleration transducers, Kyowa